On the use of the "look-locker correction" for calculating T1 values from MOLLI
View Article:
PubMed Central - HTML
To estimate true T1, a "LL correction" T1est=(/-1)T1* [Eq. 1] has been proposed... The LL correction (Eq. 1) is based on the following conditions : 1) continuous imaging; 2) spoiled-gradient-echo (SPGR) readout; 3) TR << T1; and 4) negligible initial delays after inversion (TI0) and between images (e.g., IR cine)... For cardiac imaging (T1≈1200 ms, TRR ≈1000 ms), this violates condition 3 and can also be shown to underestimate T1est... Eq. 1 thus causes T1est to illogically become a function of TI0 which leads to overestimation at longer TI0... Finally, MOLLI involves the interleaving of three LL blocks, each with an incremented TI0... The resulting composite curve consists of two distinct regions of magnetization recovery (Figure 2)... This has the counterintuitive effect of causing the composite T1* to be larger than the individual T1*s... The MOLLI acquisition does not satisfy the requirements on which the LL correction is based... Under certain conditions, this yields a reasonable estimate of T1, with the error being strongly dependent on the range of TI0... In practice, TI0 is typically too short to completely offset the effects of the LL correction, resulting in the observed systematic underestimation of T1... The use of multiple LL blocks in MOLLI was intended to improve accuracy by increasing the sampling of the relaxation curve... Instead, it can be shown that this distinguishing feature of MOLLI has the unexpected effect of essentially averaging out errors introduced by the LL correction. No MeSH data available. |
Related In:
Results -
Collection
License 1 - License 2 getmorefigures.php?uid=PMC4044058&req=5
Mentions: The LL correction (Eq. 1) is based on the following conditions [2]: 1) continuous imaging; 2) spoiled-gradient-echo (SPGR) readout; 3) TR << T1; and 4) negligible initial delays after inversion (TI0) and between images (e.g., IR cine). MOLLI uses a bSSFP readout which does not satisfy conditions 1 and 2. If the entire bSSFP readout were conceptually replaced with a single excitation pulse [3], MOLLI would approximate a LL-SPGR acquisition. However, this is not a theoretically valid simplification, and both T1* and T1est will be underestimated (Figure 1). This simplification also implies TR = TRR (RR interval). For cardiac imaging (T1≈1200 ms, TRR ≈1000 ms), this violates condition 3 and can also be shown to underestimate T1est. Further, MOLLI utilizes non-zero TI0s which violate condition 4. As TI0 increases, the curve-fitting algorithm appropriately produces an increasingly negative y-intercept A-B (thus larger B/A) and a longer T1* (because of the added true T1 relaxation during TI0). Eq. 1 thus causes T1est to illogically become a function of TI0 which leads to overestimation at longer TI0. Finally, MOLLI involves the interleaving of three LL blocks, each with an incremented TI0. The resulting composite curve consists of two distinct regions of magnetization recovery (Figure 2). This has the counterintuitive effect of causing the composite T1* to be larger than the individual T1*s. |
View Article: PubMed Central - HTML
To estimate true T1, a "LL correction" T1est=(/-1)T1* [Eq. 1] has been proposed... The LL correction (Eq. 1) is based on the following conditions : 1) continuous imaging; 2) spoiled-gradient-echo (SPGR) readout; 3) TR << T1; and 4) negligible initial delays after inversion (TI0) and between images (e.g., IR cine)... For cardiac imaging (T1≈1200 ms, TRR ≈1000 ms), this violates condition 3 and can also be shown to underestimate T1est... Eq. 1 thus causes T1est to illogically become a function of TI0 which leads to overestimation at longer TI0... Finally, MOLLI involves the interleaving of three LL blocks, each with an incremented TI0... The resulting composite curve consists of two distinct regions of magnetization recovery (Figure 2)... This has the counterintuitive effect of causing the composite T1* to be larger than the individual T1*s... The MOLLI acquisition does not satisfy the requirements on which the LL correction is based... Under certain conditions, this yields a reasonable estimate of T1, with the error being strongly dependent on the range of TI0... In practice, TI0 is typically too short to completely offset the effects of the LL correction, resulting in the observed systematic underestimation of T1... The use of multiple LL blocks in MOLLI was intended to improve accuracy by increasing the sampling of the relaxation curve... Instead, it can be shown that this distinguishing feature of MOLLI has the unexpected effect of essentially averaging out errors introduced by the LL correction.
No MeSH data available.